如果您需要准确、完整的标准引用信息，请联系我们 010-85170710 ，或者 ，请注明：标准引用与查新服务 。
非常抱歉，我们暂时无法提供预览，您可以试试： 免费下载 ASTM D6569-2005(2009) 前三页，或者稍后再访问。
pH is a measure of the hydrogen ion activity in water. It is a major parameter affecting the corrosivity and scaling properties of water, biological life in water and many applications of chemical process control. It is therefore important in water purification, use and waste treatment before release to the environment.
On-line pH measurement is preferred over laboratory measurement to obtain real time, continuous values for automatic control and monitoring purposes.
1.1 This test method covers the continuous determination of pH of water by electrometric measurement using the glass, the antimony or the ion-selective field-effect transistor (ISFET) electrode as the sensor.
1.2 This test method does not cover measurement of samples with less than 100 x03BC;S/cm conductivity. Refer to Test Method D 5128
1.3 This test method does not cover laboratory or grab sample measurement of pH. Refer to Test Method D 1293
1.4 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.5 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
Copyright ©2007-2016 ANTPEDIA, All Rights Reserved
京ICP备07018254号 京公网安备1101085018 电信与信息服务业务经营许可证：京ICP证110310号