非常抱歉，我们暂时无法提供预览，您可以试试： 免费下载 ASTM D5289-2012 前三页，或者稍后再访问。
This test method is used to determine the vulcanization characteristics of (vulcanizable) rubber compounds.
This test method may be used for quality control in rubber manufacturing processes, for research and development testing of raw-rubber compounded in an evaluation formulation, and for evaluating various raw materials used in preparing (vulcanizable) rubber compounds.
The test specimen in a rotorless cure meter approaches the test temperature in a shorter time and there is a better temperature distribution in the test specimen due to the elimination of the unheated rotor found in oscillating disk cure meters.
Several manufacturers produce rotorless cure meters with design differences that may result in different torque responses and cure times for each design. Correlations of test results between cure meters of different designs should be established for each compound tested, and for each set of test conditions.
1.1 This test method covers a method for the measurement of selected vulcanization characteristics of rubber compounds using unsealed and sealed torsion shear cure meters. The two types of instruments may not give the same results.
Note 18212;An alternative method for the measurement of vulcanization characteristics is given in Test Method D2084.
1.2 The values stated in SI units are to be regarded as the standard. The values given in parentheses are for information only.
1.3 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use.
Copyright ©2007-2016 ANTPEDIA, All Rights Reserved
京ICP备07018254号 京公网安备1101085018 电信与信息服务业务经营许可证：京ICP证110310号