非常抱歉，我们暂时无法提供预览，您可以试试： 免费下载 ASTM D5836-2008 前三页，或者稍后再访问。
Diisocyanates are used in the production of polyurethane foams, plastics, elastomers, surface coatings, and adhesives (5,6). It has been estimated that the production of TDI will steadily increase during the future years.
Diisocyanates are irritants to eyes, skin, and mucous membrane and are respiratory sensitizers. Chronic exposure to low concentrations of diisocyanates produces an allergic sensitization which may progress into asthmatic bronchitis (7,8).
The Occupational Safety and Health Administration (OSHA) has a permissible exposure limit (PEL) for 2,4-TDI of 0.02 ppm or 0.14 mg/m3 as a ceiling limit. There is no OSHA PEL for 2,6x2013;TDI (9). The American Conference of Governmental Industrial Hygienists (ACGIH) has a timex2013;weighted average (TWA) Threshold Limit Value (TLV) of 0.005 ppm or 0.036 mg/m3 and a short-term exposure limit (STEL) of 0.02 ppm or 0.14 mg/m3 for either 2,4x2013;TDI, or 2,6x2013;TDI, or for a mixture of 2,4x2013; and 2,6x2013;TDI(10).
This proposed test method has been found satisfactory for measuring 2,4 and 2,6-TDI levels in the workplace.
1.1 This test method describes the determination of 2,4-toluene diisocyanate (2,4-TDI) and 2,6-toluene diisocyanate (2,6-TDI) in air samples collected from workplace atmospheres in a cassette containing a glass-fiber filter impregnated with 1-(2-pyridyl)piperazine (1-2 PP). This procedure is very effective for determining the vapor content of atmospheres. Atmospheres containing aerosols cause TDI results to be underestimated.
1.2 This test method uses a high-performance liquid chromatograph (HPLC) equipped with a fluorescence or an ultraviolet (UV) detector (1-4). ,
1.3 The validated range of the test method, as written, is from 1.4 to 5.6 x03BC;g of 2,4-TDI and 2,6-TDI which is equivalent to approximately 9.8 to 39 ppb for 2,4-TDI and 2,6-TDI based on a 20-L air sample. The HPLC method using an UV detector is capable of detecting 0.078 x03BC;g of 2,4-TDI and 0.068 x03BC;g of 2,6-TDI in a 4.0-mL solvent volume, which is equivalent to 0.55 ppb for 2,4-TDI and 0.48 ppb for 2,6-TDI based on a 20-L air sample.
1.4 The isomers of 2,4-TDI, and 2,6-TDI, can be separated utilizing a reversed phase column for the HPLC method. Because industrial applications employ an isomeric mixture of 2,4- and 2,6-TDI, the ability to achieve this separation is important.
1.5 The values stated in SI units are to be regarded as standard. No other units of measurement are included in this standard.
1.6 This standard does not purport to address all of the safety concerns, if any, associated with its use. It is the responsibility of the user of this standard to establish appropriate safety and health practices and determine the applicability of regulatory limitations prior to use. See Section 9 for specific precautions.
Copyright ©2007-2016 ANTPEDIA, All Rights Reserved
京ICP备07018254号 京公网安备1101085018 电信与信息服务业务经营许可证：京ICP证110310号